4.Huang VS, Krakauer JW (2009) Robotic neurorehabilitation: a computational motor learning perspective. J NeuroEng Rehabil 6Google Scholar
9.Kiguchi K, Hayashi Y (2015) EMG-based control of a lower-lim power-assist robot. In: Intelligent assistive robots. Springer tracts in advanced robotics, vol 106, pp 371–383Google Scholar
10.Kawasaki H, Ito S, Ishigure Y, Nishimoto Y, Aoki T, Mouri T, Skaeda H, Abe M (2007) Development of a hand motion assist robot for rehabilitation therapy by patient self-motion control. In: Proceedings of the IEEE international conference on robotic rehabilitation (ICORR), June 2007, pp 234–240Google Scholar
13.Kyrylova A (2015) Development of a wearable mechatronic elbow brace for postoperative motion rehabilitation. Electronic thesis and dissertation repository. Paper 3019Google Scholar
14.Ferris DP, Lewis CL, (2009) Robotic lower limb exoskeletons using proportional myoelectric control. In: Proceedings of the 31st annual international conference of the IEEE EMBS, pp 2119–2124Google Scholar
15.DiCicco M, Lucas L, Matsuoka Y (2004) Comparison of control strategies for an EMG controlled orthotic exoskeleton for the hand. In: Proceedings of the 2004 IEEE international conference on robotics and automation, April 2004Google Scholar
Posts Tagged elbow
According to the United Nations (UN), by 2030 the number of people over 60 years will increase by 56 per cent, from 901 million to more than 1.4 billion worldwide . As the number of older persons is expected to grow, it is imperative that government and private health care providers prepare adequate and modern facilities that can provide quality services for the needs of older persons especially in rehabilitation centers. Implementation of robotic technology in rehabilitation process is a modern method and definitely can contribute in this policy and capable in promoting early recovery and motor learning . Furthermore, systematic application of robotic technology can produce significant clinical results in motor recovery of post-traumatic central nervous system injury by assisting in physical exercise based on voluntary movement in rehabilitation .
Even if you haven’t moved your hand and arm in years due to a neuromuscular injury or disease, it is possible the MyoPro® may be able to help you use your arm and hand again.
My Own Motion
Myomo empowers individuals with a neuromuscular condition who have lost movement in a hand and arm to perform activities of everyday life. Myomo offers the MyoPro, a myoelectric elbow/wrist/hand orthosis (powered brace) to support the weak arm and enable patients to move an impaired hand and arm again. MyoPro is the only product of its kind for people who suffer from debilitating neurological disorders such as brachial plexus injury, brain or spinal cord injury, CVA stroke, multiple sclerosis or amyotrophic lateral sclerosis (ALS).
MyoPro is covered by most commercial insurance companies in the U.S., and by the U.S. Veterans Administration – click here for more information for veterans.[…]
VISIT SITE —> Home | Myomo
[Abstract] Case Report on the Use of a Custom Myoelectric Elbow–Wrist–Hand Orthosis for the Remediation of Upper Extremity Paresis and Loss of Function in Chronic Stroke.
Introduction: This case study describes the application of a commercially available, custom myoelectric elbow–wrist–hand orthosis (MEWHO), on a veteran diagnosed with chronic stroke with residual left hemiparesis. The MEWHO provides powered active assistance for elbow flexion/extension and 3 jaw chuck grip. It is a noninvasive orthosis that is driven by the user’s electromyographic signal. Experience with the MEWHO and associated outcomes are reported.
Materials and Methods: The participant completed 21 outpatient occupational therapy sessions that incorporated the use of a custom MEWHO without grasp capability into traditional occupational therapy interventions. He then upgraded to an advanced version of that MEWHO that incorporated grasp capability and completed an additional 14 sessions. Range of motion, strength, spasticity (Modified Ashworth Scale [MAS]), the Box and Blocks test, the Fugl–Meyer assessment and observation of functional tasks were used to track progress. The participant also completed a home log and a manufacturers’ survey to track usage and user satisfaction over a 6-month period.
Results: Active left upper extremity range of motion and strength increased significantly (both with and without the MEWHO) and tone decreased, demonstrating both a training and an assistive effect. The participant also demonstrated an improved ability to incorporate his affected extremity (with the MEWHO) into a wide variety of bilateral, gross motor activities of daily living such as carrying a laundry basket, lifting heavy objects (e.g. a chair), using a tape measure, meal preparation, and opening doors.
Conclusion: Custom myoelectric orthoses offer an exciting opportunity for individuals diagnosed with a variety of neurological conditions to make advancements toward their recovery and independence, and warrant further research into their training effects as well as their use as assistive devices.
[Conference paper] ERRSE: Elbow Robotic Rehabilitation System with an EMG-Based Force Control – Abstract+References
Robotic devices for rehabilitation purposes have been increasingly studied in the past two decades and are becoming more and more diffused, due to their effective support to the traditional therapy. They allow to automate in a repeatable manner the rehabilitative exercises and to quantify outcomes, giving important feedback to the therapist. This paper deals with the design, development and preliminary characterization of a robotic system, with an exoskeleton device, for assisted upper-limb rehabilitation, in which surface EMG measurements are used to implement a force-based active and resistive control. A prototype of the system has been realized, measurements of important parameters of the motion permitted to optimize the design and preliminary tests on the control strategy were carried out.
[ARTICLE] A Neuromuscular Electrical Stimulation (NMES) and robot hybrid system for multi-joint coordinated upper limb rehabilitation after stroke – Full Text
It is a challenge to reduce the muscular discoordination in the paretic upper limb after stroke in the traditional rehabilitation programs.
In this study, a neuromuscular electrical stimulation (NMES) and robot hybrid system was developed for multi-joint coordinated upper limb physical training. The system could assist the elbow, wrist and fingers to conduct arm reaching out, hand opening/grasping and arm withdrawing by tracking an indicative moving cursor on the screen of a computer, with the support from the joint motors and electrical stimulations on target muscles, under the voluntary intention control by electromyography (EMG). Subjects with chronic stroke (n = 11) were recruited for the investigation on the assistive capability of the NMES-robot and the evaluation of the rehabilitation effectiveness through a 20-session device assisted upper limb training.
In the evaluation, the movement accuracy measured by the root mean squared error (RMSE) during the tracking was significantly improved with the support from both the robot and NMES, in comparison with those without the assistance from the system (P < 0.05). The intra-joint and inter-joint muscular co-contractions measured by EMG were significantly released when the NMES was applied to the agonist muscles in the different phases of the limb motion (P < 0.05). After the physical training, significant improvements (P < 0.05) were captured by the clinical scores, i.e., Modified Ashworth Score (MAS, the elbow and the wrist), Fugl-Meyer Assessment (FMA), Action Research Arm Test (ARAT), and Wolf Motor Function Test (WMFT).
The EMG-driven NMES-robotic system could improve the muscular coordination at the elbow, wrist and fingers.
Stroke is a main cause of long-term disability in adults . Approximately 70 to 80% stroke survivors experienced impairments in their upper extremity, which greatly affects the independency of their daily living [2, 3]. In the upper limb rehabilitation, it also has been found that the recovery of the proximal joints, e.g., the shoulder and the elbow, is much better than the distal, e.g., the wrist and fingers [4, 5]. The main possible reasons are: 1) The spontaneous motor recovery in early stage after stroke is from the proximal to the distal; and 2) the proximal joints experienced more effective physical practices than the distal joints throughout the whole rehabilitation process, since the proximal joints are easier to be handled by a human therapist and are more voluntarily controllable by most of stroke survivors . However, improved proximal functions in the upper limb without the synchronized recovery at the distal makes it hard to apply the improvements into meaningful daily activities, such as reaching out and grasping objects, which requires the coordination among the joints of the upper limb, including the hand. More effective rehabilitation methods which may benefit the functional restoration at both the proximal and the distal are desired for post-stroke upper limb rehabilitation.
Besides the weakness and spasticity of muscles in the paretic upper limb, discoordination among muscles is also one of the major impairments after stroke, mainly reflected as abnormal muscular co-activating patterns and loss of independent joint control [2, 6]. Stereotyped movements of the entire limb with compensation from the proximal joints are commonly observed in most of persons with chronic stroke who have passed six months after the onset of the stroke, during which abnormal motor synergies were gradually developed. Neuromuscular electrical stimulation (NMES) is a technique that can generate limb movements by applying electrical current on the paretic muscles . Post-stroke rehabilitation assisted with NMES has been found to effectively prevent muscle atrophy and improve muscle strength , and the stimulation also evokes sensory feedback to the brain during muscle contraction to facilitate motor relearning . It has been found that NMES can improve muscular coordination in a paralysed limb by limiting ‘learned disuse’ that stroke survivors are gradually accustomed to managing their daily activities without using certain muscles, which has been considered as a significant barrier to maximizing the recovery of post-stroke motor function . However, difficulties have been found in NMES alone to precisely activate groups of muscles for dynamic and coordinated limb movements with desired accuracy in kinematics, for example, speeds and trajectories. It is because most of the NMES systems adopted transcutaneous stimulation with surface electrodes only recruiting muscles located closely to the skin surface with limited stimulation channels . Therefore, the muscular force evoked may not be enough to achieve the precise limb motions. However, limb motions with repeated and close-to-normal kinematic experiences are necessary to enhance the sensorimotor pathways in rehabilitation, which has been found to contribute to the motor recovery after stroke . Furthermore, faster muscular fatigue would be experienced when using NMES with intensive stimuli, in comparison with the muscle contraction by biological neural stimulation .
The use of rehabilitation robots is one of the solutions to the shortage of affordable professional manpower in the industry of physical therapy, to cope with the long-term and labour-demanding physical practices . In comparison with the NMES, robots can well control the limb movements with electrical motors. Various robots have been proposed for upper limb training after stroke [12, 13]. Among them, the robots with the involvement of voluntary efforts from persons after stroke demonstrated better rehabilitation effects than those with passive limb motions, i.e., the limb movements are totally dominated by the robots . Physical training with passive motions only contributed to the temporary release of muscle spasticity; whereas, voluntary practices could improve the motor functions of the limb with longer sustainability [10, 14]. In our previous studies, we designed a series of voluntary intention-driven rehabilitation robotics for physical training at the elbow, the wrist and fingers [14, 15, 16, 17, 18]. Residual electromyography (EMG) from the paretic muscles was used to control the robots to provide assistive torques to the limb for desired motions. The results of applying these robots in post-stroke physical training showed that the target joint could obtain motor improvements after the training; however, more significant improvements usually appeared at its neighbouring proximal joint mainly due to the compensatory exercises from the proximal muscles [15, 17]. In order to improve the muscle coordination during robot-assisted training, we integrated NMES into the EMG-driven robot as an intact system for wrist rehabilitation [16, 19]. It has been found that the combined assistance with both robot and NMES could reduce the excessive muscular activities at the elbow and improve the muscle activation levels related to the wrist, which was absent in the pure robot assisted training . More recently, combined treatment with robot and NMES for the wrist by other research group also demonstrated more promising rehabilitation effectiveness in the upper limb functions than pure robot training . However, most of the proposed devices are for single joint treatment, and cannot be used for multi-joint coordinated upper limb training. Furthermore, the training tasks provided by these devices are not easy to be directly translated into daily activities. We hypothesized that multi-joint coordinated upper limb training assisted by both NMES and robot could improve the muscular coordination in the whole upper limb and promote the synchronized recovery at both the proximal and distal joints. In this work, we designed a multi-joint robot and NMES hybrid system for the coordinated upper limb physical practice at the elbow, wrist and fingers. Then, the rehabilitation effectiveness with the assistance of the device was evaluated by a pilot single-group trial. EMG signals from target muscles were used for voluntary intention control for both the robot and NMES parts.
The NMES-robot system
Continue —> A Neuromuscular Electrical Stimulation (NMES) and robot hybrid system for multi-joint coordinated upper limb rehabilitation after stroke | Journal of NeuroEngineering and Rehabilitation | Full Text
[ARTICLE] Effects of repeated vibratory stimulation of wrist and elbow flexors on hand dexterity, strength, and sensory function in patients with chronic stroke: a pilot study – Full Text PDF
[Purpose] The aim of this study was to investigate the effects of repeated vibratory stimulation to muscles related to hand functions on dexterity, strength, and sensory function in patients with chronic stroke.
[Subjects and Methods] A total of 10 stroke patients with hemiplegia participated in this study. They were divided into two groups: a) Experimental and b) Control, with five randomly selected subjects for each group. The experimental group received vibratory stimulation, while the control group received the traditional physical therapy. Both interventions were performed for 30 minutes each session, three times a week for four weeks.
[Results] There was a significant within-group improvement in the box and block test results in both groups for dexterity. Grip strength improved in both groups but the improvement was not statistically significant.
[Conclusion] The vibratory stimulation activated the biceps brachii and flexor carpi radialis, which increased dexterity to grasp and lift the box and block from the surface. Therefore, repeated vibratory stimulation to muscles related to hand functions improved hand dexterity equality to the traditional physical therapy in patients with chronic stroke.
In the rehabilitation training and assessment of upper limbs, the conventional kinematic model treats the arm as a serial manipulator and maps the rotations in the joint space to movements in the Cartesian space. While this model brings simplicity and convenience, and thus has been overwhelming used, its accuracy is limited, especially for the distal parts of the upper limb that execute dexterous movements.
In this paper, a novel kinematic model of the arm has been proposed, which has been inspired by the biomechanical analysis of the forearm and wrist anatomy. One additional parameter is introduced into the conventional arm model, and then both the forward and inverse kinematic models of five parameters are derived for the motion of upper arm medial/lateral rotation, elbow flexion/extension, forearm pronation/supination, wrist flexion/extension and ulnar/radial deviation. Then, experiments with an advanced haptic interface have been designed and performed to examine the presented arm kinematic model. Data analysis revealed that accuracy and robustness can be significantly improved with the new model.
This extended arm kinematic model will help device development, movement training and assessment of upper limb rehabilitation.
Body joint movement analysis is extremely essential for health monitoring and treatment of patients with neurological disorders and stroke. Chronic hemiparesis of the upper extremity following a stroke causes major hand movement limitations. There is possibility of permanent reduction in muscle coactivation and corresponding joint torque patterns due to stroke . Several studies suggest that abnormal coupling of shoulder adductors with elbow extensors and shoulder abductors with elbow flexors often leads to some stereotypical movement characteristics exhibited by severe stroke patients . Therefore continuous and effective rehabilitation therapy is absolutely essential to monitor and control such abnormalities. There is a substantial need for home-based rehabilitation post-clinical therapy.