Posts Tagged grasping
Stroke is among the top three causes of death in the United States, but nothing comes close to stroke as the leading cause of long-term disability. After patients survive a stroke, their risk of having another stroke increases, along with their likelihood of suffering a serious disability as a result. However, medical and technological advances have made it easier to help patients cope and recover. Occupational therapy is an effective way to restore mobility and reduce future risks for stroke survivors.
Therapy for stroke survivors often involves “re-training” or reprogramming the brain after neurological damage. As we learn more about the relationship between the brain, muscles, and connective tissue, one stimulating innovation is emerging as a top tool for recovery. Today, many patients are relying on a stroke rehabilitation gloves & dynamic splints to reverse damage, restore mobility, and reduce pain after a stroke.
But how, exactly, does wearing these orthoses treat symptoms of stroke survivors? Truth is, there are many benefits for patients who incorporate a glove or a dynamic splint into their recovery process.
Problems Stroke Can Cause
Especially with strokes, survivors can suffer from impaired function, weakness and spasticity. Spasticity causes involuntary muscle contractions in the arms and can even cause even short-term or long-term paralysis as the tendons and tissues around the muscles get tighter.
Strokes can really affect upper arm movements too. Survivors only use their affected upper limb approximately 3 hours per day.
Individuals who have not suffered a neurological injury use their dominant hand for an average of 9 hours per day. Patientsless than 14 days following stroke use their affected upper limb only 38 minutes out of a 9-hour day.
Shortening of muscles and connective tissue can start occurring within hours/days. Maintaining a shortened position for a prolonged period of time leads to fibrous adhesion formation, loss of sarcomeres and a loss of tissue extensibility.
There Is Hope After A Stroke
Fortunately, we can respond to spasticity, and lessened arm movements and muscle tone by harnessing the brain’s own plasticity. Cortical Plasticity, also known as neuroplasticity, is the brain’s remarkable ability to reorganize itself by forming new neural connections based on individual experiences, lifestyle and environment. It essentially is the brain’s ability to “re-program” itself through mass practice, task-oriented arm training.
To get these neuroplastic changes, patents participate in skill-dependent rather than simply use-dependent activities. Skill-dependent activities are specific and progressively challenging tasks whereas use-dependent activities are repetition tasks in the absence of a meaningful challenge or an activity that requires problem solving strategies.
With these skill-dependent activities cortical maps are continuously remodeled throughout life and after injury by experiences and learning in response to activity and behavior from the stroke survivor. Stimulated through this task training, the brain has the ability to reorganize and form new connections between the intact neurons. The healthy surrounding tissue takes over some of the functions of the damaged area of the brain.
This Is Where Stroke Rehabilitation Gloves and Dynamic Splints Come Into Place
Task-specific training with rehabilitation gloves and dynamic splits improve upper extremity function in individuals suffering from neurological injuries. Treatment options are limited for neurological clients who cannot effectively incorporate their hand for functional grasp and release activities. This is where dynamic splints can really help rehabilitation.
Dynamic Splints Help Train the Brain
If the hand and arm muscles are no longer functional, it’s especially important to re-learn basic functions first, such asgrasping and releasing objects. A stroke rehabilitation device like the SaeboFlex can make this process easier for some patients and possible for those who otherwise would have no function left.
For a vast majority of stroke survivors, especially ones with incomplete spinal cord injury, patients do not exhibit sufficient active wrist and/or finger extension to allow the hand to be functional. Stroke recovery gloves like SaeboFlex has the biomechanical advantage in allowing prehension grasp and release activities for individuals with moderate to severe hemiparesis.
The SaeboFlex and other rehabilitative dynamic splints actually step in to compensate for some of the patient’s biomechanical disadvantages.
The majority of patients with neurological or spinal cord damage cannot extend their fingers or move their wrists, but this orthosis imitates the hand’s natural functions and makes it possible to grasp and release objects. The goal is to make the hand functional again, but it also minimizes joint damage and pain.
Dynamic Splints Help Fight Contracture
When stroke survivors lose function in their upper limbs after a stroke, sometimes hard static splints are used to keep the arm and wrist in a “neutral” position and avoid muscle contracture. Unfortunately, some studies have shown that static splinting is ineffective against muscle contracture, and others have actually linked the practice to joint damage and contracture. Contracture is a loss of motion over time due to abnormal shortening of the soft tissue structures spanning one or more joints. These include skin, ligaments, tendon, muscles and joint capsules.
The ideal splint is dynamic, moveable and helps stretch out muscles, tendons and ligaments like the SaeboStretch.
The splint’s energy-storing technology allows individuals suffering from spasticity to stretch comfortably and safely resulting in increased motivation and compliance. It allows the fingers to move through flexion caused by associated reactions and increased tone.
A dynamic splint may prevent contracture after a stroke, as well as:
- Reduce joint pain
- Protect the joints
- Prevent edema (buildup of excessive fluid in the muscle tissue)
- Allows the fingers to move through flexion caused by postural changes associated reactions and increased tone.
- Gradually repositions the fingers into extension
As patients recover from a stroke, every effort to restore strength and function is invaluable. Using a stroke rehabilitation dynamic splint is a proven way to reduce pain and complications while survivors focus on their recovery. It may also open up new possibilities by restoring the use of their arms.
Stroke Recovery Glove For Improved Hand Functionality
It’s important to keep the muscles active after a stroke, in order to prevent stiffness and shortening of the tissue. Activity also helps to keep pathways between the brain and muscles open. Stroke recovery gloves that promote sensorimotor stimulation are useful to stroke survivors for many different reasons, from preventing complications to making life-changing therapy methods possible.
Therapy is a big part of the recovery process after a stroke, and occupational therapy often incorporates basic elements such as towels or small objects as patients learn to grasp, release, hold, and perform other basic tasks. Stroke rehabilitation gloves like the SaeboGlove can help with these activities. The SaeboGlove is a functional stroke recovery hand glove that has a tension system integrated into it which helps individuals extend their fingers and thumb after grasping. This helpful stimulation helps the neuroplastic changes the brain needs to help reprogram itself.
Positive results have been witnessed and experienced with professionals and clients when integrate them into upper arm rehabilitation exercises.
Results of stroke rehabilitation gloves in therapy have included:
- Significant increases on the Fugi-Meyer Assessment and Box and Block Test, which are designed to test the elbow’s control and strength during reach-to-grasp tasks
- Reduced jerkiness of the wrist, shoulder, and elbow joints during reach-to-grasp therapy tasks
- Improved flexion and abduction
- Support for hand and finger extension after loss of mobility
- Increased motor recovery
- Increased grip strength
- Active improvement to the overall functionality of patient hands, with some enjoying nearly full functionality
- Strong potential for future improvement of arm/wrist mobility.
Improved Stroke Recovery With Gloves And Dynamic Splints
Skill-dependent physical activities have long helped stroke survivors reprogram their brains, strengthen their muscles, and improve their quality of life after neurological damage. Stroke rehabilitation gloves and dynamic splints can give the patients the needed stimulation and help they are needing to progress. They are great to help protect the joints while improving strength and mobility. Because patients can incorporate these gloves and dynamic splints into occupational therapy as well as everyday tasks, they make it easier to achieve independence during their stroke recovery.
[Abstract] Design and Test of a Closed-Loop FES System for Supporting Function of the Hemiparetic Hand Based on Automatic Detection Using the Microsoft Kinect Sensor
[Abstract] A soft robotic supernumerary finger and a wearable cutaneous finger interface to compensate the missing grasping capabilities in chronic stroke patients
[Abstract] Design and Test of a Closed-Loop FES System for Supporting Function of the Hemiparetic Hand Based on Automatic Detection using the Microsoft Kinect sensor
[ARTICLE] The Robotic Sixth Finger: a Wearable Compensatory Tool to Regain Grasping Capabilities in Paretic Hands – Full Text PDF
Among the most promising field of applications of wearable robotics there are the rehabilitation and the support in activities of daily living (ADL) of impaired people. In this paper, we propose two possible designs of a robotic extra-finger, the Robotic Sixth Finger, for grasping compensation in patients with reduced hand mobility, such as post-stroke patients. The idea is to let the patients be able to grasp an object by taking advantage of the wearable device worn on the paretic limb by means of an elastic band. The Robotic Sixth Finger and the paretic hand work jointly to hold an object. Adding a robotic opposing finger is a promising approach that can significantly improve the grasping functional compensation in different typologies of patients during everyday life activities.
Wearable robots are expected to work very closely, to interact and collaborate with people in an intelligent environment . Traditionally, wearable robotic structures have been mainly used in substitution of lost limbs (e.g., prosthetic limbs) or for human limb rehabilitation (e.g., exoskeletons). However, the progress in miniaturization and efficiency of the technological components is allowing more light and compact solutions, enhancing user’s safety and comfort, while opening new opportunities for wearable robot use . Together with exoskeleton and prosthesis, a very promising research direction seems to be that of adding robotic limbs to human, rather than substituting or enhancing them . This addition could let the humans augment their abilities and could give support in everyday tasks to impaired people. This paper investigates how to compensate the capabilities of the human hand, instead of developing additional robotic extra-arms, as discussed for instance in . The idea of using an extra-finger to support the human hand in grasping functions was initially proposed in . Then, independently both in  and [7, 8], the authors proposed the use of extra fingers to support the human hand to grasp objects whose size does not fit a hand or in executing bimanual tasks with one hand. The main difference is that in [7, 8], the goal was to minimize the size and the weight of the unique extra limb, while in , two extra fingers were used so to hold objects. While in  the authors developed a control strategy to grasp and manipulate objects, in  the authors mainly focused on the use of extra fingers for post-stroke patients.
Focusing on the hand, many wearable devices have been proposed in the last decade, especially for hand rehabilitation and function recovery. A review on robotassisted approaches to motor neurorehabilitation can be found in . In  the authors presented a comprehensive review of hand exoskeleton technologies for rehabilitation and assistive engineering, from basic hand biomechanics to actuator technologies.
However, most of the devices proposed in literature are designed either to increase the functional recovery in the first months of the rehabilitation therapy, when biological restoring and reorganization of the central nervous system take place, or are designed to augment human hand capabilities of healthy subjects by coordinating the device motion to that of the hand.
To the best of our knowledge, only few works target on the robotic compensation of hand function in the latter phase of rehabilitation. This means that patients usually after 6–9 months of rehabilitation must rely only on compensatory strategies by improving adaptations that increase the functional disparity between the impaired and the unaffected upper limb .
This work focuses on the compensation of hand function in patients with paretic limbs, e.g. chronic stroke patients. The final aim is to provide the patient with an additional robotic finger worn on the wrist. The Robotic Sixth Finger is used together with the paretic hand to seize an object, as shown in Fig. 1. The systems acts like a two-finger gripper, where one finger is represented by the Robotic Sixth Finger, while the other by the patient paretic limb. The proposed device goes beyond exoskeletons: it adds only what is needed to grasp, i.e. an extra thumb. We presented in  a preliminary version of a robotic extra-finger showing how this wearable device is able to enhance grasping capabilities and hand dexterity in healthy subjects. In , we also presented an object-based mapping algorithm to control robotic extra-limbs without requiring explicit commands by the user. The main idea of the mapping was to track human hand by means of dataglove and reproduce the main motions on the extra-finger. This kind of approach is not suitable for patients with a paretic limb due to the reduced mobility of the hand. Therefore, we developed a wearable interface embedded in a ring to activate and use the finger .
In this work, we propose two possible designs of devices. The first model is a modular fully actuated finger. The other design consists of an underactuated finger which is compliant and consequently able to adapt to the different shapes of the objects.
For validation purposes, pilot experiments with two chronic stroke patients were performed. The experiments consisted in wearing the Robotic Sixth Finger and performing a rehabilitation test referred to as Frenchay Arm Test [15, 10]. Finally, we present preliminary results on the use of the extra-fingers for grasping objects for Activities of Daily Living (ADL).
[Abstract] The Soft-SixthFinger: a Wearable EMG Controlled Robotic Extra-Finger for Grasp Compensation in Chronic Stroke Patients. – IEEE Xplore
This paper presents the Soft-SixthFinger, a wearable robotic extra-finger designed to be used by chronic stroke patients to compensate for the missing hand function of their paretic limb. The extra-finger is an underactuated modular structure worn on the paretic forearm by means of an elastic band. The device and the paretic hand/arm act like the two parts of a gripper working together to hold an object. The patient can control the flexion/extension of the robotic finger through the eCap, an Electromyography-based (EMG) interface embedded in a cap. The user can control the device by contracting the frontalis muscle. Such contraction can be achieved simply moving his or her eyebrows upwards. The Soft-SixthFinger has been designed as tool that can be used by chronic stroke patients to compensate for grasping in many Activities of Daily Living (ADL). It can be wrapped around the wrist and worn as a bracelet when not used. The light weight and the complete wireless connection with the EMG interface guarantee a high portability and wearability. We tested the device with qualitative experiments involving six chronic stroke patients. Results show that the proposed system significantly improves the performances of the patients in the proposed tests and, more in general, their autonomy in ADL.
[ARTICLE] Kinematic analysis of upper limb motion: Feasibility, preliminary results in controls and hemiparetic subjects, prospects
The aim of this study is to develop a valid and standardized instrumental analysis of upper limb (UL) motion in stroke patients.
Sixteen controls and 15 hemiparetic subjects (mean age = 54 ± 18,2 years old; Fugl-Meyer Upper Limb 41,4 ± 12,4) underwent kinematic motion analysis (passive markers, Optitrack) of pointing and grasping tasks. We examined the ability to perform a single pointing task and three reach-to-grasp tasks: key turning, reaching and grasping a can, reaching and grasping a cube; at a self-selected speed and as fast as possible. Speed, accuracy and efficiency of each movement were quantified and compared between controls and hemiparetic subjects, and between the ipsilateral of control subjects and the affected side; to describe reaching and grasping.
For reaching, movement time of hemiparetic UL was longer, less smooth (peak velocity, jerk), less direct (higher index path ratio) and associated with more trunk compensation (higher trunk/hand ratio). Movement time, jerks and trunk/hand ratio were the most discriminant variables between hemiparetic UL and ipsilateral/control UL, in any task analysed. Trunk displacement was greater in grasping than in reaching tasks. For grapsing tasks, movement time is the most discriminant factor between hemiparetic and control/ipsilateral UL, especially for the key turn task. Movement alterations were also found for ipsilateral limb. Association between kinematic variables and clinical features during reaching time (Fugl-Meyer, MAL, WFMT, ARAT) was greater for the task “grasping a can”.
Our results are similar to those of the literature, but suggest that we have to privilege some of the most relevant kinematic parameters. This standardization phase emerging after a validation phase of the techniques can make the biomechanical analysis of the upper limb as easy and valid as gait analysis and should help to develop the quantified measurement of prehension. This protocol is currently in process to objectively assess the therapeutic effects of rehabilitation treatments (botulinum toxin, induced constraint therapy).
[ARTICLE] A full upper limb robotic exoskeleton for reaching and grasping rehabilitation triggered by MI-BCI
In this paper we propose a full upper limb exoskeleton for motor rehabilitation of reaching, grasping and releasing in post-stroke patients. The presented system takes into account the hand pre-shaping for object affordability and it is driven by patient’s intentional control through a self-paced asynchronous Motor Imagery based Brain Computer Interface (MI-BCI). The developed antropomorphic eight DoFs exoskeleton (two DoFs for the hand, two for the wrist and four for the arm) allows full support of the manipulation activity at the level of single upper limb joint. In this study, we show the feasibility of the proposed system through experimental rehabilitation sessions conducted with three chronic post-stroke patients. Results show the potential of the proposed system for being introduced in a rehabilitation protocol.